The summit of evolution
Premium class ultrasound performance is no longer just for color system. a full-feature, premium black & white ultrasound system.
- Synthetic Aperture
- Synthetic aperture control is a method for overcoming the limits of the physical channel in software methods. As shown in the diagram, to create one scan line, TX and RX work is performed twice.
- Harmonic Imaging
- The basic theory of harmonic imaging is that body tissue reflects ultrasound signals at twice the frequency of the scanning frequency, or the second harmonic with scanning frequency. Harmonic imaging is an advanced version of conventional harmonic imaging.
- Full Spectrum Imaging
- Spectrum imaging is designed for imaging by considering the entire frequency territory that the probe supports, no matter how the TX frequency is designated. The frequency range can be broadened for better penetration and resolution.